CMU200 LABVIEW DRIVER DOWNLOAD
You need to change it to true to initialize the devices. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. How to select the proper driver for your specific environment is described in this chapter. If you do not query the instrument when you open it, the firmware is treated as 0 and all subsequent commands will error with an insufficient firmware revision error message. Hi you can try this.
|Date Added:||20 March 2012|
|File Size:||55.40 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
After the hardcopy is configured and triggered in line Customer response 5 2.
Moxa Device Manager 2. The device descriptor you used “GPIB In case of using a file as storage destination, a destination path and a destination file name has to be defined. The file created by this step can be downloaded via the remote control connection Source code availability The complete source code of the following examples is available on the instrument specific driver download site.
LXI features, More information.
On top of labvied, it can be used as a full-featured More information. Also the file on the instrument will be deleted afterwards. Kindly clarify me regarding this. This is shown in Figure 3.
CMU error – Discussion Forums – National Instruments
In the application note 1MA In the source code below line 13 specifies which file will be loaded from the instrument and stored on the host PC. I’ve already installed the RSFSex intrument library that i found in the web site.
cju200 You can also try to assign a different address to the device. This can be done in two steps: Configuration Requirements for ipc Operation The hardware and operating system of the Labvuew must comply with a list of minimum requirements for proper operation with the ipc. System Requirement and Connection More information. Furthermore a subset for transferring files from the instrument to the host PC is available allowing the transfer of screenshot images to the host PC.
If you do not need to use secondary address, you don’t have to use any value for this part. General hints When using the instrument drivers, it is crucial to set a valid hardcopy file path and file name.
The information in this guide is subject to change without notice. B Operating Manual System Requirement and Connection It does not matter what I do to the signal, I can make the target time of my chirp ten million and make my simulation final time 1 billion, unless my window length is 1, I get the same error. We cannot be held liable for any problems arising from.
INSTR” seems to be the default value used in the driver vis, is the primary address of the instrument really 20 and secondary address 0? Converters 1 Contents Overview and Features We will not contact the customer unless you tell us to.
Rohde & Schwarz CMU Universal Radio Communication – Instrument Driver – National Instruments
My problem is that every time I run this vi I get an error telling me that my window length must be less than my signal length but greater than zero. This is shown in Figure 6. Line 28 distinguishes between an instrument error or an error which recognized by the driver.
It is useful for instance in case of 6 incorrect external trigger when the instrument does not 7 respond to any other command because of waiting for 8 lavview. Both methods are simple ways to establish communications with your instrument. Once these settings are complete, a copy of the screen can be initiated.
State of Michigan Data Exchange Gateway. Message 2 of 9.